Lens Flame Retardant Testing Machine
Lens Flame Retardant Testing Machine meets the standard: QB
to determine whether the material has the required flame retardant
properties to ensure the safety of users.
Lens flame retardant testing machine test instrument requirements:
can heat the regular cylindrical steel bar (φ6 * 300mm) resistance
Flammability tester with adjustable maximum temperature of 800 ° C,
equipped with pneumatic device
Automatically raise and lower the heating rod to prevent the tester
from touching the heating rod.
A. Determine the test location (different materials should be
B. Place the test sample
C. Start the instrument and put down the heating rod when the
temperature of the heating rod rises to 650 degrees Celsius.
The test sample is contacted for 5 seconds, and then the heating
rod is removed;
D. Observe whether the tested part of the sample continues to burn.
1. Thermocouple: φ6*300mm
2. Automatic pressure control
3. Microelectronic automatic timing
4. Size: 35 × 40 × 45 CM (length × width × height)
5. Weight: 25 Kg
6. Power: AC220V 50Hz