300KV Ceramic tube penetrant 40mm Directional Radiographic
If the unit get out of control in the states of exposure and aging,
such as over-voltage, under-voltage, over-current and
under-current,etc. voice will prompt,the system will turn off high
voltage automatically and return to ready state,in the meantime
corresponding protective indicator lamp will be on and indicate the
type of troubles.Press"OFF"key for clearing the trouble after you confirm the trouble type,
and then try it again.
When the tube head temperature is too high and overtemperature
indicator (OV)is flashing, now all keys are unworkable,exposure can be taken
again when the temperature return to normal and the unit return to
ready state automatically.
If time stops and displays in confusions,power supply must be cut
off immediately and check up the voltage,gas pressure of the
generator,air humidity and cables connection,after informing it
being normal, the machine can be turned on again.
XXG3005 Ceramic tube orientation/directional
The HUATEC Flaw detector products have been widely used for
nondestructive testing in spaceflight industry, welding engineering
industry, shipbuilding industry, boiler pipe industry,
petrochemical engineering industry, aircraft manufacturing,
maintenance industry, automotive industry, defense industries, etc.
Main technical parameters
|Max Penetration thickness Steel (A3)||50mm|
|Film Type||TianJin Ⅲ|
Double lead foil Intensifying screen
|Input||Voltage (V)||220V±10% 50―60Hz|
|Output||Tube voltage (KVP)||150-300|
|Tube current (mA)||5|
|Operation Mode||Intermittent operation, Radiating time to cooling time is 1:1|
|Circuit Structure||Micro-computer control ,Modular structure,Voice prompt|
|Cooling||Force air cooling|
|Insulation||SF6 Gas Insulation|
This machine is composed of intelligent controller-ray generator,
power cable and connecting cable.
1.The controller will consist of the following sections:
Front Panel: LED display, status and failure indicator, touch keyboards (see
the introduction of control panel)
Side Panel: power cable socket and connecting cable socket, power switch,key
switch, fuse socket, warning lamp and wire connecting terminal of
Internal Circuit: power circuit, rectifying circuit, filter circuit, inverter
circuit, control circuit and other circuited.
2.The x-ray generator will consist of the following sections:
X-ray tube, High voltage transformer, Temperature relay, Gas
pressure gauge and Cooling fan etc.
3.The schematic diagram of the complete machine:
|Directional glass X-ray tube||1 Pcs|
|Generator||1 Set|| |
|High voltage joins cable||1 Pcs||25 Meters|
|Controller||1 Set|| |
|Power cable||1 Pcs||10 Meters|
|Grounding cable||1 Pcs||5 Meters|
|Accessories||1 Set|| |
|(Including: alarm lamp,1 Pcs; lead letter, 1 box; Film dark
cassettes 5 pcs; intensifying screen 5 pcs; Wire penetrameter (IQI)
1 set; fuse1 pc)|