Tomy automatic microprocessor control model ES 315 made in Japan
Once turned on the system automatically sets and displays a pre-set
value. Operation is done simply by pressing the start switch.
ES-315/ES-215 enables the set temperature and time for each mode
program to be stored in its memory.
MODE/PROCESS display enables real time monitoring of operation
ES-315 offers three mode programs.
Sterilization Mode It contains a delay start timer function that
allows operation to start 1 to 99 hours later.
Sterilization/warmingMode This mode is useful when not removing
culture media from the autoclave immediately after sterilization.
Sterilized media can be kept warm to prevent coagulation.
HeatingMode This mode program automates the dissolution of culture
media that can sometimes be a painful process when using a water
bath to boil them.
Chamber having a flat inner surface
All models are supplied with stainless steel wire mesh baskets. No
protrusions on inner surface of the chamber enable smooth loading
and unloading of the basket as well as block design of all pipe
fittings ensures safe operation.
Front access for easy maintenance
The exhaust bottle to capture and store exhaust steam is placed on
the front panel of the autoclave unit. Access to the chamber for
replacing water inside, as well as for routine maintenance and
inspection, is done on the front of the unit, which also permits
operators to replace bellows, saving costs for services by the
A microprocessor controlled monitoring system provides safe
Incomplete sterilization protection
In case of abnormal chamber temperature rise, the autoclave stops
its operation. In case of abnormal decrease in temperature, the
sterilization timer is temporarily suspended and then reactivated
when the set temperature is reached.
Chamber overpressure protection
If overpressure in the chamber is detected, the autoclave stops all
the function and indicates an error message simultaneously.
Chamber interior overtemperature protection
If the temperature inside the chamber exceeds the set temperature
by 5°C or more, the autoclave stops all the function and indicates
an error message simultaneously to avoid overheating.
Chamber exterior overtemperature protection
The thermostat switch detects the temperature of chamber exterior
wall, which allows the autoclave to stop all the function if the
temperature exceeds 130°C.
With following specifications:
Autoclave also feature three cycles of operation and Graphic
Built-in Steam exhaust bottle to Collect Steam pressure to prevent
it to go out the instrument
Inside the chamber overheat prevention, Outer wall of the chamber
overheat prevention, Overpressure prevention, Temperature sensor
disconnection prevention, Empty heating prevention, Leakage
breaker, Safety valve
Malfunction prevention devices
Low water level detection, Exhaust valve knob open/close detection,
Insufficient sterilization detection,
Chamber lid open/close detection
an Automatic Air Release Time